Forums › SM Series Discussions › SM200A RF Input Connector
- This topic has 1 reply, 2 voices, and was last updated 5 years ago by Andrew.
mikesParticipantmikes March 27, 2018 at 2:41 pm
I’m curious about the choice of connector for the RF input on the SM200A. Is it just a normal type-N connector or a precision type-N connector? It appears the typical type-N connector is rated for use up to 11 GHz, and the precision type-N connector is rated up to 18 GHz. I know the laboratory grade connectors used on some of the “big box” analyzers can get quite expensive, but I was wondering how much of a difference in actual RF performance there is and what the reason for using N on the SM200A was (beyond being more economical, although that’s a big reason).
AndrewModeratorAndrew March 27, 2018 at 3:23 pm
The bulkhead connector is from Centric RF, part number C3614. It is considered a precision type-N connector. Particularly good VSWR. Data sheet link below. It is spec’d to 18GHz and in our tests fine up to 20GHz. The main reason for using the type-N was due to customer feedback, mainly revolving around robustness, rigidity, and prevalence in labs.
Let us know if you have additional questions.
You must be logged in to reply to this topic.