hhonza501Participanthhonza501 June 12, 2018 at 6:39 am
Hi, I bought the second BB60c and when compared I found that the original one (about one year old) has a significantly lower sensitivity. Same setting Spike, same antenna and computer, fault?
AndrewModeratorAndrew June 14, 2018 at 5:38 pm
Since each BB60C has different calibration data which controls the sensitivity of the receiver based on the reference level, it is possible for the same reference level to provide different gain/attenuation settings. A good test would be to set the reference level to -50dBm, match RBW/VBW/center freq and measure the level of the noise floor (trace averaging can help). You can also manually control the gain/atten by enabling the manual gain/atten settings (file menu).
Feel free to email email@example.com with any more information or pictures.
You must be logged in to reply to this topic.