BINDINIRParticipantBINDINIR March 30, 2018 at 11:49 am
Why is “phase noise measurement mode” not available on BB60C?
Thanks a lot.
Justin CrooksModeratorJustin Crooks April 2, 2018 at 9:53 am
Because of the BB60C architecture, its phase noise, especially at low frequencies (e.g. below 1 GHz) is not very good. So we decided to disable phase noise measurements rather than provide poor results.
The SM200A is recommended for phase noise measurements inside 100 kHz offset. The SA44B can make pretty good phase noise measurements as well, especially outside 100 kHz offset.
You must be logged in to reply to this topic.