Ivan Ionov November 16, 2015 at 1:30 am
- This topic was modified 6 years, 9 months ago by Ivan Ionov.
I usually collect several traces in SNA, record one trace, disable update of this trace, record next…
Then I switch Spike to Sweep mode, and all traces with unchecked “Update” remains on screen.
Also, I see simple bug – when I switch from Sweep to Zero span, center frequency always resets to 1GHz. Other modes remember center frequency.
Attachments:You must be logged in to view attached files.
AndrewModeratorAndrew November 16, 2015 at 8:17 am
Thanks for the heads up on these. The Zero Span center frequency resets to 1 GHz each time you start the program. If you click the zero-span button on the control panel, it will use the current center frequency as the new center frequency for zero-span. Also, if you use the file menu to get to zero-span, it will use the last center frequency that was set in zero-span. Hope this helps.
You must be logged in to reply to this topic.