BDick August 14, 2022 at 3:02 pm
- This topic was modified 1 year, 3 months ago by BDick.
I am having problems getting the Limit Line to display on my sweep. I have entered limit line points, Enabled the limit line, and clicked on Show Line, but nothing is displayed. See attached
I am using Spike 3.7.3 with a BB60C
Attachments:You must be logged in to view attached files.
AndrewModeratorAndrew August 15, 2022 at 8:48 am
The limit line amplitudes you listed are in positive dBm. I suspect you meant for them to be negative? (-50, …) Right now the limit line is being drawn off the top of the screen.
You must be logged in to reply to this topic.