thientran November 12, 2019 at 5:53 pm
- This topic was modified 4 years ago by thientran.
I have 1 SA44B and I want to use it to measure the amplitude of the signal at the center frequency.
I used the function (in the API):
AmpliFreqCenter = myMeasRcvr.RFAmplitude .
The data receiving speed is very slow (about 500 ms / 1 sample). How to improve the speed?
AndrewModeratorAndrew November 13, 2019 at 6:18 am
The API you are currently using has been deprecated and we unfortunately don’t support it anymore. Several years ago we released a newer API for the SA44B which you can download in our SDK at the link below.
Using the newer API, I would configure a sweep and use the sweep data to measure the amplitude at a given frequency. You can use the Spike software to determine the sweep settings needed for your measurement and use them when setting up your sweep in the API. There is a sweep example you can use to get you started.
There is no measurement receiver function in the new API, but based on your short description, it does not sound like you need that. It sounds like a simple sweep would work.
Let me know if you have follow up questions.
thientranParticipantthientran November 19, 2019 at 7:35 pm
Thank you for your help.
I have done it and succeeded
You must be logged in to reply to this topic.