Ivan IonovParticipantIvan Ionov November 16, 2015 at 1:44 am
I connect input to output, do store thru calibration and look at a trace. In <10MHz range the trace is unusable at all – it jumps from -30dB to +2dB, on one reading it can be +2db, on second reading it can be -20db…
First picture with min-max hold.
Second picture with 3 traces – min, max and average.
I don’t think it’s a hardware issue, Span mode works correctly in this range.
Attachments:You must be logged in to view attached files.
Justin CrooksModeratorJustin Crooks November 16, 2015 at 9:23 am
Thank you for bringing this to our attention. It looks like there is a software bug we will need to address. I’ll let you know what we find.
You must be logged in to reply to this topic.