Technical Software and Hardware Change Log

Please send bug reports and feedback to support@signalhound.com

- Software Version 1.0.13 (6/2/2026)
* Added support for Maury Microwave uCal Automatic cal modules.

- Software Version 1.0.12 (5/26/2026)
* Updated USB driver.
* Added spur reject setting to help improve spurious performance in specific measurement scenarios.

- Software Version 1.0.11 (4/27/2026)
* Added the ability to scale the smith chart through the trace reference level.
* Added SCPI command for exporting SnP files.
  :SENSe<ch>:EXPort:SNP <filename>, <port>
* Added SCPI commands for setting the reference level, division, and reference level position.
  - :CALCulate<ch>:MEASure<tr>:Y[:SCALe]:RLEVel
  - :CALCulate<ch>:MEASure<tr>:Y[:SCALe]:DIVision
  - :CALCulate<ch>:MEASure<tr>:Y[:SCALe]:RPOSition

- Software Version 1.0.10 (2/9/2026)
* Added support for sliding loads.
* Mkr->Ref Mkr now enables the ref marker if it wasn't already enabled.
* Added more precision to statistics printouts.
* Fan settings are now remembered and used the next time the application is launched.
* Updated product numbers for Signal Hound system polynomial calibration kits.
* Added SCPI commands for minimizing/maximizing the application window.
  - :DISPlay:MINimize
  - :DISPlay:MAXimize
* Bug fix: Fixed Mkr->Start/Stop/Center implementations.
* Bug fix: Marker updates should now function properly on smith plots.

- Software Version 1.0.9 (10/2/2025)
* Allow arbitrary frequency spacing for sNp files used in databased cal kits.
* Better auto scale behavior for LogMag and SWR trace formats.
* Added trace scale coupling.
* Added SCPI commands for
  - Trace auto scale
  - Trace auto scale all
  - Trace default scale
  - Trace default scale all
  - Trace scale coupling
  - Display hide

- Software Version 1.0.8 (8/26/2025)
* Traces now update during a calibration. After measuring a standard, traces associated with the channel being
  calibrated will update. This allows you to verify correct connections during calibration.
* Improved screenshots with option to omit control panels.
* Added new Signal Hound system calibration kit definitions.
* Added ability to import calibration kits.
* Added ability to export SNP data files from databased cal kit standards.
* Added SCPI commands for setting low pass time domain compatible settings.
  - :CALCulate<ch>:MEASure<tr>:TRANsform:TIME:SLPFrequency
  - :CALCulate<ch>:MEASure<tr>:TRANsform:TIME:SLPFrequency?
* Added preference for disabling the standard selection dialog when only one standard matches the class being measured.

- Software Version 1.0.7 (7/24/2025)
* Improved measurement accuracy when the calibration is interpolated across the hardware band breaks.
* Added SCPI commands for saving and loading instrument state.
  - :SYSTem:PRESet[:USER]:SAVE <filename>
  - :SYSTem:PRESet[:USER]:LOAD <filename>
* Bug fix: Fixed marker peak left.
* Bug fix: Markers should now update properly when changing measurement configurations.

- Software Version 1.0.6 (6/11/2025)
* Fixed "User Manual" file menu entry. Was not properly opening the user manual.

- Software Version 1.0.5 (5/12/2025)
* Added marker peak tracking and corresponding SCPI commands.
  - :SENSe<ch>:MEASure<tr>:MARKer<mkr>:FUNCtion:TRACking OFF|MAXimum
  - :SENSe<ch>:MEASure<tr>:MARKer<mkr>:FUNCtion:TRACking?
* When selecting a calibration kit,
  - The calibration kit selected now defaults to last used kit.
  - User calibration kits are now listed before system kits.
* When adding a new standard to a calibration kit, the corresponding class may optionally be automatically assigned to it.
* Improved UI feedback for:
  - Selecting traces on plot.
  - Performing automatic port extension measurements.

- Software Version 1.0.4 (4/9/2025)
* Official public release.

- Software Version 1.0.3 (2/25/2025)
* Added Ohm marker readout in the configuration
  - Time domain enabled
  - Lowpass step response selected
  - Trace format is real
* Minimum sweep size is now 1 point.
* Minimum step size is now 0Hz.
* Bug fix: Fixed impulse step response for time domain measurements when multiple extrapolated points are needed.
* Bug fix: Fixed marker coupling for Smith chart plots.
* Bug fix: Fixed marker coupling across channels with different stimulus.
* Bug fix: Time domain x-axis fixed for different transform types.
* Bug fix: Step size stimulus control now working properly.

- Software Version 1.0.2 (2/6/2025)
* Added the following SCPI commands.
  - :SENSe<ch>:MEASure<tr>:DATA:RDATA?
  - :SENSe<ch>:MEASure<tr>:DATA:RECR?
  - :SYSTem:PREFerence:FCAL[:STATe]

- Software Version 1.0.1 (2/5/2025)
* Hide gender selections/text for thru standards.
* Added support for the following SCPI commands.
  - :SENSe<ch>:CORRection:CSET:ACTive?
  - :SENSe<ch>:CORRection:CSET:TYPE?
  - :SENSe<ch>:CORRection:CSET:ETERm?

- Software Version 1.0.0 (2/3/2025)
* Pre-production release.
